|(2) Operating Procedure :|
|1)You must prepare for the measurement of the
Active Hydrogen (H - one atom of hydrogen) and Molecular Hydrogen
(H２- two atoms of hydrogen) :
Firstly, you just place one piece of our “ Active Hydrogen Water Generator H-01”
into a plastic water bottle. Then just before measuring, you should shake the
plastic water bottle about 20 times in order for the particles to accumulate near
the top of the inside of bottle.
|2) Operation is very simple.|
|(1) When you switch it on, a red LED display will
then count down by seconds.
After precisely 60 seconds, when the instrument reaches 0.00,
the instrument is ready for use, and you can conduct the measurement immediately.
|(2) After 60 seconds, you shake the plastic bottle about 20 times.
Then you place the brass cap of a detector into the top of the opened plastic bottle.
Then, a red LED starts flashing until it reaches the right position, at which point it stops flashing.
(3) Then you can read the concentration of active hydrogen (H) and molecular hydrogen (H2) by mg/litre :
up to 3 digits.
|(4) Switch off|
|3) Re-setting instructions :|
|(1) Re-set to 30 seconds.
When you press the reset button, a red LED display will then count down by seconds.
After precisely 30 seconds, when the instrument reaches 0.00, the instrument is ready for use,
and you can conduct the measurement immediately.
You can repeat the measurement test every 30 seconds by pressing a reset button.
|(2) Switch off|
|A 3 digit red LED display indicates the
concentration of active hydrogen (H) and molecular hydrogen (H2)
level from 0.00 upwards, in increments of 1mg/liter.
The maximum should be 2.44 mg/litre with saturated hydrogen.
When you measure, be careful not to get the sensor wet.
When a sensor gets wet, it will become out of order.
A sensor is incorporated just inside of the brass cap.
Before measuring, please pour away water about 5cm from the top
of the plastic water bottle, in order for you to avoid getting the sensor wet.